|Todays radiation graph|
|Last 7 day's radiation graph. The high peak was caused by an hardware error (faulty connector causing noise, extra counts)|
Browsing the internet, I found some relatively cheap ex-army radiation detectors at an army-dump shop. One of them appeared to be suitable to even detect the (usually low) background radiation levels: A Frieseke & Hoepfner FH40T Geiger counter (fitted with a FHZ76V energy-compensated geiger-mueller tube), sensitive to γ (gamma) radiation and β (beta) radiation over 0.25MeV.The FHZ76V tube actually contains a Valvo 18550 tube, which is equivalent to Centronics ZP1320, Mullard Mx164 and LND-713 (found in this Probe Selection Guide and here)
The specs of the ZP1320 tube claim a sensitivity of 9cps/mR/h for Cs-137 (540cpm/mR/h). For 'normal' background (0.025-0.045mR/h) this results in a counting rate of approx.10-20cpm.. Where I live, I measure values varying between 4cpm up to 25cpm. This variation is caused by the randomness of the decay of radioactive elements.